Insitu Raman Setup for Deep Ocean Investigations Applying Two 1000 m Optical Fiber Cables and a 785 nm High Power Diode Laser

A laboratory Raman-set-up is used to demonstrate the feasibility of deep ocean measurements using 1000 m long low OH optical fib res for the excitation of the substances under study and the collection of the Raman photons. As excitation laser a 785 nm broad area distributed feedback (DFB-BA) high-power d iode laser with an optical power of 1.25 W and a spectral emission width below 0.6 nm suitable for Raman spectroscopy was used.After passing beam shaping and focusing optics, a 1000 m long fiber, and a pressure-resistant optode275 mW of optical power were available at the sample. The optode also collects the Raman photons and focuses the light into the second 1000 m long fib re, which transfers the light into a spectrometer. To determine the reachable signal-to-noise ratio, Raman spectra were taken from natural minerals (sulfates, carbonates, sulfur). Even for weak Raman-signal and short integration times of 0.5 s per spectrum, the signal-to-noise ratio of 60 would be suitable for the in-situ Raman spectroscopy in deep ocean applications. Even longer optical fibres would increase to integration time up to 60 s. The shown set -up offers the opportunity to used different laser sources and spectrometers on board of a ship, only the fibres and the optode had to be suitable for deep sea conditions.


Introduction
Since decades considerable attention has been given to Raman spectroscopy as a reliable and effect ive tool for remote sensing and in-situ applications in various fields e.g. ocean science. Vital to this process are continous developments in laser sources (compact diode lasers) and high performance CCDs. Due to its non-invasive and non-destructive character and the fact that water produces only weak Raman signals in the fingerprinting range,Raman spectroscopy is introduced as an important technique for potential investigation of geochemical materials on the sea floor.
Due to the special environmental conditions in the deep sea, co mprisinghigh pressures and extreme temperatures, measurements are often restricted in taking samples followed by laboratory analyses.However, insitu investigations are desirable in order to monitor the specimensin their ambient surroundings. For that purpose, a pressure-compensatedspectrometer system for deep-ocean investigations has been tested successfully on a remotely operated vehicle ROV for measurements o f gas hydrat es [1], [2] and at hyd rothermal andcold seep systems [3]. Furthermore, hydrothermal and cold seep minerals have been averifiedusing a laboratory setup [4] and a deep sea Raman probe for measurements of pore water geochemistry was developed recently [5]. Nevertheless, immersion of the whole Raman equip ment in the deep sea has some limitations. In that way, it is difficu lt to use large laser system and spectrometers with h igher sensitivity improving the spectral qualitydue to the space-limited pressure-resistant housings.
In earlier worksit has been demonstrated that fiber optic probing for remote Raman spectrometry over a few meters has several advantages for many application areas [6][7][8]. In that context, Myrick and Angel [9] showed that it was possible to perform Raman measurements using a 100 m optical fiber for conventional Raman spectroscopy and a 250 m fiber for surface enhanced Raman scattering. Additionally, a long fiber-optic remote Raman probe with a length of 100 m for the detection and identification of weak Raman scatterers was discussed by Schoen et al. [10].
All of these studies applied Kr + and Ar + lasers as excitation light sources using emission lines in the visible spectral region. Th is choice offers a limitation, due to the increased fluorescence interference, e.g. by impurities in natural substances, which can co mp letely obscure the Raman signals of the sample. To reduce this undesired fluorescence,the excitation wavelength can be shifted to the near-infrared region, e.g. at 785 n m. Fu rthermore, this excitation wavelength offers the advantage of being outside of the water absorption band which limits the application of Fourier transform IR instruments operating at 1064 n m [11]. After successfully realizing a 785 nm d istributed feedback diode laser [12] with 200 mW of optical output power, a 1.5 W high power broad area (BA) d istributed feedback (DFB) diode laser emitting at 785 n m [13] was developedrecently. Due to its small spectral bandwidth belo w 0.6 n m (10 cm -1 ), this laser is well suited as excitation source for Raman spectroscopy.
In this work,we present a combination of the high power BA DFB and two 1000 m optical fibers in order to realizea laboratory Raman setup suitable for insitu experiments over long distances. This approach provides the possibility to separate the marin ized optical sensor head from the on-board component containing laser and spectrometer. Hence, different laser sources and high-resolution spectrometers on board of the ship can be applied and different kinds of targets like minerals, fluids, and gases can be probed by exchanging the optical sensor head according to the measurement task. In addition, investigations in space limited areas like drilled holes are possible by mounting the optical sensor head on a flexib le arm.
Beside deep ocean usage,the presented experimental configuration might be also of interest forfurther applications where a spatial separation of spectrometer and laser fro m the target is necessary. This co mprises harsh (high pressure, high or low temperatures, aggressive or explosive targets) or space-limited (e.g. boreholes)measurement conditions.

Laser Source
For Raman measurements under deep sea conditions using long excitation and collection fibers ,the light sources must fulfill two preconditions. On the one hand, the spectral line width of the laser source has to be adapted to the typical bandwidth of the substances under study as well as to the spectral resolution of the spectrometer. In our case, the targets are sub-marine minerals present in the deep sea environment typically having a spectral width of characteristic Raman bands of 10 cm -1 [14]. In co mbination with the spectrometer resolution of 10 cm -1 this requires a laser emission width of 10 cm -1 or below, corresponding to a spectral width of  = 0.6 n m at a wavelength of  = 785 n m.
On the other hand, the output power fro m the diode laser has to be largeenough to ensurean excitation power ex fiber of at least 50 mW to enable reasonable measurement times . For this reason, standard ridge waveguide (RW) DFB lasers cannot be used asthey have typical maximal output powers of about 200 mW as reported by Wenzel et al. [12]. Taking into account the typical attenuation of fibers in this spectral range, a min imu m output power of 1 W is required for deep sea applications. Hence, the capabilit ies of a broad area (BA) DFB diode laser offer the potential to meet these requirements. The laser devicehas a stripe width of 50 µm and a length of 1.5 mm. It was grown with a t wo-step metal organic vapor phase epitaxy and the internal grating for frequency stabilizat ion was manufactured by holographic exposure. More details about the device structure and the manufacturing processcan be found in Maiwald et al. [13]. The diode laser is mounted p-side down on a C-mount. The power-voltage-current characteristic at an operat ional temperature of T= 25℃ is shown in Figure 1 exh ibiting a threshold current ofI th = 290 mA and a slope efficiency of S = 0.85 W/A. The maximu m optical output power at an injection current of I = 2 A amounts to P = 1370 mW.  Figure 2 shows the spectral tuning of the DFB BA laser fro m the threshold at 290 mA up to 2 A (i.e. 1370 mW ). It can be seen that the laser spectrally tunes from 783.1 n m to 784.5 n m. Although at injection currents larger than 400 mA Two 1000 m Optical Fiber Cables and a 785 nm High Power Diode Laser two modes occur, the spectral width in the whole working range does not exceed 0.5 n m. Hence, in any case the spectral emission width remains below the maximum acceptable value of 0.6 n m. Accordinng to its output power and spectral width, the BA DFB laser is well-suited as excitat ion light source for Raman measurements through long optical fibers.
To design a proper optical focusing system, the beam quality of the device was measured. Figure 3 shows the lateral near field (top) and far field (bottom) p rofiles at output powers of 1000 mW and 1250 mW at anoperation temperature of 25℃. The beam widths with 95 % power content at the near field are about 48 μm and 46 μm at P = 1000 mW and P = 1250 mW , respectively. The far field angle measured at 95 % power content is 14° at 1000 mW and amounts to 15° at 1250 mW. The full width at half maximu m (FWHM) o f the vertical far field profile is 22°. This allows a highly efficient coupling of the BA DFB diode laser output beam into an optical fiber. For the Raman setup, a low OH optical fiber cable (AS 105/125) with a length of 1000 m was used to transfer the excitation laser radiation to the samp le. According to the manufacturers specifications ,the fiber has an absorption of about 42 % at 785 n m [15]. Due to the fiber d iameter of 100 μm and a nu merical aperture of NA = 0.22, the maximu m collection angle of the fiber is 25.4°. Based on the laser beam characteristics,we have used the simu lation program winA BCD [16] to determinesuitable lenses to efficiently couple the laser beam into the optical fiber. As a result, an aspherical lens with f = 8 mm was applied to collimate the laser beam and a lens with f = 16 mm was used to focus the beam into the fiber. Using this configuration and according to the near field and far field profiles, we expect spot widths of about 96 μm and 92 μm at 1000 mW and 1250 mW, respectively. The convergence angles of the focused beam are expected to be 9° at 1000 mW and 10° at 1250 mW.
To prevent the laser diode from back-reflections, an optical isolator (Gsä nger DLI-1) was used. Applying alaser output power of 1250 mW , an optical power of665 mWwas obtained after the isolator correspondingto apower loss of about 47 % caused by the isolator. After passing the 1000 m optical fiber cable , we obtained 345 mW , i.e. the losses due to the coupling and the attenuation of the fiber at 785 n m are 48 %. To determine the coupling losses into the fiber, measurements with a fiber length of 1 m were carried out.   In order to measure the Raman signal originating fro m the fiber itself, the laser beam was coupled into the fiber using the configuration mentioned above and the output radiation fro m the fiber was collimated by a lens with f = 16 mm. Subsequent application of two Raman edge filters rejects the laser light and only transmits the Raman Stokes radiation which was then focused into the spectrometer by another lens with f = 16 mm. The recorded Raman spectrum of the optical fiber is presented in Figure 4 (a). Here, pro minent silica bands are dominant in the range up to 800 cm -1 and could overlap with sample Raman signals, thus disturbingthe measurement. Therefore, a 785 n m band pass filter with a transmission of 91 % at 785 n m and a bandwidth of 8 n m at FWHM was placed after the excitation fiber to efficiently reject the Raman scattering radiation arising fro m the quartz fiber (see Figure 4 (b)).

Experi mental Raman Setup
In order to transfer the excitation laser rad iation to the target and the backscattered Raman signal to the detector,two low OH fiber optic cab les (AS 105/ 125) with a length of 1000 m each, provided by FiberTech GmbH, Germany, were used.
The laboratory setup, which should prove the capability for deep sea Raman investigations, is schematically shown in Figure 5. The output beam of the 785 n m diode laser (1) with an optical power of1250 mW is collimated by an aspherical lens (all lenses were purchased from ThorLabs, Germany) (2) and then passes through an optical isolator (Gsä nger DLI-1, 60 d B) (3) to prevent back-reflections into the laser diode. Using a lens with f = 16 mm (4), the laser beam is coupled into the1000 m excitat ion fiber (5). A lens (6) with f = 16 mm serves for co llimat ion of the output beam fro m this excitation fiber. A band pass filter (Quarterwave, Germany) (7) was used to spectrally clean the laser radiat ion and to eliminate the frequency shifted radiation arising fro m Raman scattering inside the 1000 m quartz fiber. The laser beam is reflected by a dielectric mirror (ThorLabs, Germany) (8) to a Raman edge filter (Quarterwave, Germany) (11) which guides the radiation to an achromatic lens (9) with f = 8 mm. This lens serves for focusing the laser beam onto the sample (10) as well as for collecting the backscattered radiation. In this case, the focused beam has a spot diameter of 50 μm and the convergence angle is equal to 28°. On ly the Raman Stokes radiation passes through the two Raman edge filters, while the Ray leigh and anti-Stokes co mponentsare blocked. By means of an achro mat ic lens (12) with f = 16 mm, the Raman Stokes radiat ion is coupled into the 1000 m collection fiber (13) with a spot width of 100 μm and a convergence angle of 14° resulting in a high efficientcoupling. By means ofthis fiber, the Raman scattered radiation of the sample is transferred into the spectrometer (Princeton Instrument PI 320) (14) and the spectra were recorded with a back-illu minated deep-depletion CCD (DU420-BR-DD, Andor) (15) thermo-electricalycooled down to an operation temperature of -60℃.
The optical components which have to be integrated into a pressure-resistant optode for deep ocean measurements are included in the dashed rectangle in Figure 5. A prototype of such a device was already developed by our group and tested up to a pressure of 20 MPa which is equivalent to a water depth of 2000 m [17].
As mentioned above, we obtained a laser power of 345 mW before the optodeusing the BA DFB diode laser and a low OH fiber as excitation fiber.In that way, the optical power at sample position amounts to 275 mW which was sufficient large to realize short measuring times of 0.5 s for one Raman spectrum. For the investigationsof the minerals, 10 spectra were averaged each for 5 different points on the sample.

Materials
To test the capability of our setup for deep sea investigation, selected natural mineral samples as sulfates which are produced by hydrothermal vents, carbonates which can be found in shells, and sulfur which is produced by bacteria [18], were used (see Tab. 1). The samples are provided by the geoscientific collection of the Department of Geological Sciences, University of Bremen.

Data Treatment Two 1000 m Optical Fiber Cables and a 785 nm High Power Diode Laser
To obtain Raman spectra suitable for the identification and to study the minerals, the fluorescence background from all Raman spectra has to be removed. Fo r this purpose, we used SOLIS software (Andor), wh ich records the data and subtracts the backgroundof the spectra usinga cubic spline fitting. Characteristic points of the signal background were selected to obtain a continuous baseline curve wh ich was then subtracted from the Raman data resulting in an essentially fluorescence-freeRaman spectrum. As an examp le for this data treatment, the unprocessed Raman spectrum of anhydrite is displayed in Figure 6 (A).Here, the dotted line refers to the baseline curve indicating a high signal background. Thus e.g. the very weak anhydrite Raman signal at 499 cm -1 is masked. After baseline correction,majo r and minor anhydrite signals can be recognized in Figure 6 (B), e.g. the small band at 676 cm -1 and the strong bands at 1019 cm -1 and 1131 cm -1 . Hence, direct intensity calculation of the Raman signals can be done after baseline correct ion.
Beside the characteristic Raman signals, theanhydrite spectrum in Figure 6 exh ibits several peaks above 1200 cm -1 which originate fro m lu minescence. Previous studies have shown that the presence of Mn and rare earth elements could produce these luminescent bands in natural anhydrite [4,19].

Sulfates
Anhydrite (CaSO 4 ), gypsum (CaSO 4 · 2H 2 O), and barite (BaSO 4 ) which include the sulfate ion (SO 4 2-) within their structure (see Table 1) were used for our investigations. The Raman spectraof these minerals are characterized by specific vibrations of the sulfate ion. Due to structural differences of these materials, slight energy shifts of the peak position can be recognized in their Raman spectral signature. Figure 7 exhibits the Raman spectra of anhydrite, gypsum, and barite. According to these spectra, the minerals can be identified with data fro m the literature [20]. The Raman signals of these three sulfate minerals are reproducible and the signal positions as well as their vibrational assignments are indicated in Figure 7. Here, the minor Raman bands can be recognized with signal-to-noise ratios ranging from 20 to 60. Hence, applying a 1000 m optical fiber,measurement times of only0.5 s enablesufficient signal-to-noise ratiosof the Raman signals suitable fo r detection and identification purposes of these mineral species . To co mpare our study with the previous study from White 2009[4], the relative intensities of minor Raman bands with respect to the major Raman signals were calculated. In that way, the relative intensity values were in the range fro m 0.03 to 0.33 wh ich is in good agreement with White`s study.

Carbonates
As examp les of carbonates , we selected aragonite and calcite, which are two poly morphs of calciu m carbonate. Despite oftheir identical chemical co mposition (CaCO 3 ), both minerals have different crystal structures (see Table 1), which cause slight differences in their Raman spectra. Figure  8 presents the Raman spectra of both carbonates together with the corresponding assignment of their vibrat ional modes [21]. The signal-to-noise ratios of the minor Raman bands vary fro m 20 to 44, i.e. these small signals can be clearly detected after transmission through the 1000 m optical fiber. The relative intensities of the minor Raman signals range from 0.01 to 0.1 wh ich is in good conformance with the data presented in the literature [4] .

Sulfur
Elemental sulfur in the S 8 configuration is also a molecule of great interest which can be found in the deep ocean. Filamentous sulfur is produced by sulfur-o xid izing bacteria [18] which provide a base for the chemosynthetic food web in the deep sea. Due to the transmission characteristics of our Raman edge filter with a cut-on wavelength at 400 cm -1 fro m the laser line, only two Raman signals of sulfur in the region between 400 cm -1 and 500 cm -1 can be observed whereas the signals located in the 20 -300 cm -1 region are blocked. Hence, in Figure 9 t wo Raman signals due to S-S stretching can be identified at 434 cm -1 and 472 cm -1 . The minor Raman band (434 cm -1 ) can be detected with a signal-to-noise ratio of 35 and its relative intensity amounts to 0.04. The results of our study using the low OH optical fiber fo r Raman signal collect ion are in good accordance with the literature data [3], [4].

Esti mation for Deep Sea Experiments
Based on our investigations using 1000 m optical fibers , we can estimate the experimental conditions for deep sea measurements requiring longer optical paths. Appling e.g. a 4 t imes longer optical path the integration t ime necessary to record spectra of co mparab le quality increases to about 16 s.Regarding connection losses , integration times of less than 1 min would occur, still enabling in-situ deep-sea measurements in the presented configuration.

Conclusions
We presented a Raman setup combining a 1.25 W high power BA DFB d iode laser and two 1000 m low OH fiber optical cables (AS 105/125)suitable for insitu experiments over long distances. One optical fiber was used to transfer the excitation laser light to the sub-marine mineral sample, whereas the second fiber serves to transfer the back-scattered Raman Stokes radiation fro m the sample to the spectrometer. Both fibers are connected to a specially designed optoderealizing optical powers of 275 mW at the sample position and enabling efficient spectral filtering.
With this setup, Raman spectra of selected sulfates, carbonates, and sulfur can be detected and identified with extreme short measurement times of 0.5 s only which are well suited for in-situ deep sea investigations. Even small Raman bands of the minerals can be recognized with signal-to-noise ratio up to 60 showing the capability of the configuration with two 1000 m optical fibers and a BA DFB diode laser for in-situ deep sea investigations.
For a further imp rovementof the signal-to-noise ratio,it would be interesting to incorporate other Raman techniques like shifted excitation Raman difference spectroscopy. For such experiments diode laser with high optical power within two slightly shifted excitation lines are required.